DTM-3107 LCD waterproof thermometer with detachable probe Product ID: DTM-3107
LCD waterproof thermometer with detachable probe
◎ with 280mm long stainless steel probe.
◎ replacement, demolition stainless steel probe
◎ The maximum and minimum temperature memory function.
◎ low battery warning.
◎ power switch.
◎ accuracy can be displayed to the nearest one.
◎ with numeric lock key.
DTM-3107 Thermometer Application: Suitable for cooking,
conditioning, laboratories, pools, environmental measurement
and other uses.
-50°C ~ +150°C OR -58°F ~+302°F
-50°C ~ +300°C OR -58°F ~+572°F
2.Resolution: 0.1° for -19.9° ~ +199.9°, otherwise 1°
3.Accuracy : ±1°C in the range of -30°C ~ +150°C
±1.8°F in the range of -22°F ~ +302°F
±1°C in the range of 151°C ~ +250°C, otherwise more than ± 2°C
±1.8°F in the range of 303.8°F ~+482°F, otherwise more than ± 4°f
4.Ambient temperature for plastic case: -10°C ~ +50°C or +14°F ~+122°F
5.Battery: 1 x 1.5 volts g13 size or equivalent
To obtain the best results. The probe tip should be immersed at least 10mm. Into the tested substance.
Remove the battery from the battery compartment if the thermometer will not be used for a long duration of time or during storage.
The thermometer cannot be used or put inside an oven or microwave.
DTM-3107 Probe Thermometer Specifications:
Measuring range: -50 ℃ ~ 300 ℃
Working environment: 0 ℃ ~ 50 ℃
Accuracy: ± 1 ℃ (-30 ~ 250 ℃) other ranges 2 ℃
Display Update: 1 second
Probe Specifications: 280mm stainless steel
Battery: button battery (357A) X 1
First, connect the thermometer probe and tighten the probe to avoid seams seepage.
Press [NO / OFF] button to turn on the power outlet if the battery symbol is on the screen, the battery
power is low
Please replace the battery, to avoid a large measuring error occurring.
Maximum and minimum reading settings:
Pressing [MAX / MIN] button appears on the display panel shows the current MAX is the maximum
the temperature of the measured
Degree reading, press the [MAX / MIN] button, the screen displays MIN, and as the lowest temperature
Leave the maximum and minimum readings:
The maximum and minimum reading mode, pressing [MAX / MIN] button for 5 seconds to leave the
maximum and minimum mode
If the pH is too high for measuring body measurement, measurement should be cleaned after the probe.
When replacing batteries, remove the screw on the screw should pay attention to cushion can not be lost.
A long time without measurements, remove the batteries to avoid poor quality battery discharge the phenomenon is not normal,
Cause leakage damage thermometer.
Thermometer recommended every year must be returned to parity check, once to ensure the accuracy of the
The panel, key Description:
1 Celsius / Fahrenheit switch
2. Reading lock
3 Max / Min display reading
4 Power switch
5. Reading display