TIG-320 Infrared Thermal Imaging Camera Product ID: TIG-320
Infrared Thermal imager
TIG-320 infrared thermal imager adopts the latest technology, which can merge visible light image and infrared thermal image together. This infrared imaging device is advanced and professional, accurate and efficient to provide better quality and high-quality thermal image.
Built-in 3GB memory, check on computer or other equipment, widely used in fire fighting, archeology, transportation, agriculture, electronics manufacturing and other fields
Temperature Ranges: -20 ~ 300 °C ( -4 ~ 572 F°)
Image Resolution: 320 x 240
Accuracy: ±(2%) or +2°C
Viewing angle / Focusing Distance: 56° x 42° / 0.15 meter
Wavelength Range: 8 ~ 14μm
Screen Update Rate: 9Hz
Screen: 3.2" full viewing angle TFT display
Display Color Palette: Rainbow / Iron Red / Cool / White Hot / Black Hot
Emissivity (ε): 0.01 ~ 1.00 (Adjustable)
Storage Capacity: 3G
Auto Power Off: 5 minutes / 20 minutes / no automatic shutdown
Working Environment: 0 ~ 45°C / lower than 80%RH Non-Condensing。
Storage Environment: -20 ~ 60°C / lower than 70%RH Non-Condensing。
Transmission Cable: Micro USB 2.0
Power : Rechargeable lithium battery(#18650)
Dimension: 90 × 105 ×223mm
Accessory: Waterproof protective cover, instruction manual.
Can this camera work in a total darkness? TECPEL Infrared Camera is used for detecting infrared radiation from object which won't be affect by the strength of any visible light. It performs in darkness as well as under sunshine.
Why the temperature I get from the camera is far below the real data it should be? Different material and same material with different surface may be different in IR emissivity. To adjust the IR emissivity according to the table in our manual is important so that you would be able to get the most accurate temperature results.
Why I can see 2 images not align with each other? When you are in IR-VL mode, the visible light image and infrared rays image are all on the monitor. You may need to calibrate the image while the detecting distance is changed. Please refer to the Image calibration instruction in the manual.
Can this model TIG-320 detect electrical power lines or hot water pipe behind the walls? That's really depends on the scenario. Generally, you can't detect the object behind a barrel such as a wall that does not allow infrared rays going through. But in most of scenario we can indirectly ‘see’ the hot water pipe behind a wall as it heats up the wall and generate typical thermal distribution on the surface of a wall.
What is the farthest distance it can detect? The best test distance is 20 cm-50 cm which covers most of professional test scenario.